The most common complications in tibial shaft fracture are:
Malalignment with malunion
Nonunion is the result of a combination of biological and/or mechanical errors that interfere with the bone healing process. In the tibial shaft there are only muscles in the lateral side. Therefore, the vascular supply of the medial tibia is poor. This can sometimes lead to nonunion. Improper handling of the muscles and soft tissue in combination with poor surgical technique and suboptimal implant selection can lead to tibial nonunion.
Osteomyelitis can occur in the tibia in presence of an open fracture or iatrogenic contamination.
4. Malalignment with malunion
Rotational malalignment in comminuted fractures can occur due to a lack of recognizable landmarks that aid the surgeon in proper alignment of the limb. Mild loss of length or a moderate malalignment on the sagittal plane (procurvatum or recurvatum) does not affect the patient’s functional outcome, whereas malalignment on the frontal (varus or valgus) or axial plane can severely compromise limb function.
Limb alignment can be assessed by clinical evaluation and intraoperative fluoroscopy or radiographs.
5. Pin too long
The radiograph shows a pin inserted too far caudal into the stifle joint (left) / tarsal joint (right).
6. Plate too short
Lateral radiograph of a 42-B3 tibial fracture stabilized with a short 3.5mm “string of pearl” plate. Note the sagittal plane angulation.
Early post-operative failure of the repair by tibial fracture. The failure is likely the result of a combination of a short plate and the weakening effect of the most proximal screw placed through or near the caudal cortex.
7. Plate failure
When a plate is too short it can lose the biomechanical advantage of bridging the bone, and may lead to a catastrophic implant failure.
Preoperative radiographs of a 42-B2 fracture of the tibia in a 14-year-old Border Terrier.
The fracture was repaired using a 2.0/2.7 Veterinary Cuttable Plate. Although the plate was intended to work as a neutralization plate, the lack of reduction and stabilization of the short oblique fracture (arrow) make this plate a bridging plate.
Implant failure by fatigue at 3 weeks postoperative at the level of the previously highlighted short oblique fracture.